Q&A: Two pH Sensors – what is causing this difference?
January 2019 Newsletter
The data shown above originate from two different pH sensors deployed in the same well-mixed test bath. Both sensors were cleaned, tested, and calibrated before deployment, and were given 24 hours to equilibrate before recording data. So why does each sensor appear to be offset from the other, and why does the sensor shown in green appear to have gaps in data?
Find the Answer
These results highlight two noteworthy differences between each sensor technology:
- Initial Accuracy: the SeapHOx V2 has an initial accuracy of ±0.05 pH or better, while most glass electrode pH sensors (the HydroCAT-EP included) have an initial accuracy of ±0.1 pH. The figure above demonstrates each sensor’s accuracy with the shaded area around each line (light green = HC-pH accuracy, light red = SeapHOx V2 accuracy, yellow= overlap). Based on the validation samples, both pH sensors were reporting data within their accuracy specifications, but the SeapHOx V2 was clearly more accurate from start to finish.
- Sampling resolution: the “stepwise” changes in the HC-pH data are the result of lower resolution. The sensor is unable to resolve changes lower than 0.01 pH, so small changes in pH aren’t recorded until a large enough change can be detected. The SeapHOx V2 is able to resolve small changes of up to 0.004 pH, resulting in smooth lines as pH drops.
While the data from the ISFET pH sensor are superior in accuracy, resolution, and stability, there are still many applications where glass electrode sensors such as the one integrated with the HydroCAT-EP are sufficient. However, when you need top of the line pH for long-term moored deployments, an instrument using an ISFET pH sensor (SeaFET V2, SeapHOx V2, or Deep SeapHOx V2) is necessary.
Did you get a chance to read about the HyperNAV program that we support in partnership with NASA? Sea-Bird Scientific is proud to be part of the NASA PACE Mission, which contributes to the UN Ocean Decade's 7 Outcomes. On December 5, 2017, the United Nations declared...
A New Paradigm for Ocean Color Satellite Calibration The HyperNAV program is part of the NASA PACE (Plankton, Aerosol, Cloud, Ocean Ecosystem) Mission, set to launch in 2024. The NASA PACE Mission is an initiative aimed at helping scientists understand how carbon...
Celebrating and honoring our LGBTQIA+ communities At Sea-Bird Scientific, we are proud to stand with members of the LGBTQIA+ community during Pride Month 2022. This year, you may have noticed we changed our logo on social media to feature a rainbow throughout the...