Salinity Spikes: Sensor or Environment?
These data show a subtle problem that is producing “fuzzy” salinity and oxygen data. While the overall shape of the profile appears reasonable, small spikes are present throughout the CTD cast. The black lines encompass a particularly strange spot where both salinity and oxygen jump away from their respective averages. Is there a problem with the equipment, or are these spikes a result of the environment?
Find the Answer
Answer: This CTD’s pump was not functioning properly, resulting in a poor response from the conductivity and oxygen sensors. This problem is most pronounced while travelling through gradients or during times of significant ship heave.
These data originated from the secondary temperature/conductivity/oxygen channels on a 9plus CTD. The figure above shows data from the primary sensors superimposed over the secondary sensor data (red is primary salinity, green is primary oxygen). The primary sensors had a fully functional pump, so the data are much cleaner, with tighter grouping between samples and greater similarity in the upcast and downcast.
Repairing the pump associated with the secondary Temperature/Conductivity/Oxygen plumbing on the 9plus fixed the problem, and data from the secondary sensors matched nicely with data from the primary sensors.
Did you get a chance to read about the HyperNAV program that we support in partnership with NASA? Sea-Bird Scientific is proud to be part of the NASA PACE Mission, which contributes to the UN Ocean Decade's 7 Outcomes. On December 5, 2017, the United Nations declared...
A New Paradigm for Ocean Color Satellite Calibration The HyperNAV program is part of the NASA PACE (Plankton, Aerosol, Cloud, Ocean Ecosystem) Mission, set to launch in 2024. The NASA PACE Mission is an initiative aimed at helping scientists understand how carbon...
Celebrating and honoring our LGBTQIA+ communities At Sea-Bird Scientific, we are proud to stand with members of the LGBTQIA+ community during Pride Month 2022. This year, you may have noticed we changed our logo on social media to feature a rainbow throughout the...