Q&A: Regular Spikes in Salinity
The salinity profile shown above was from an arctic deployment in rough seas. At a certain point, we see regular salinity spikes of about ± .01 psu in the upcast data.
What’s unusual is the regularity and size of each spike – electrical issues may manifest with a consistent pattern, but we’d typically expect those to show extreme errors in data. Can you determine what caused this?
Find the Answer
Answer: Sampling Errors from Ship Heave
“Ship heave” is the rocking motion of the vessel, often the result of rough seas. For a CTD deployed from a sea cable, this rocking motion leads to pulling and slackening of the sea cable when the ship rocks in each direction. The resulting pull/slack causes deceleration/acceleration of the underwater package. As the instrument decelerates, it entrains water within the package that can continue downward past the sensors. This water is of different temperature and conductivity than the water at the bottom of the package, and it causes a sampling error.
You can see this in action by plotting “Descent Rate” (meters depth per second) alongside salinity. Note how the spikes in salinity correspond to “loops” in descent rate. In calm waters, descent rate is typically consistent at around 1-2 meters/second. But descent rate varies in rough seas. In extreme conditions, the pull on the sea cable can actually cause the CTD to ascend during the descent, resulting in a “loop”
With most winch / A-frame hardware, these sampling errors are unavoidable. Knowing how to recognize them is crucial. Learn how to filter them out in SBE Data Processing’s “Loop Edit” module.
Hear from Dr. Kim Martini, Senior Oceanographer, Erin Keen, Field Sales Support, and Laura Collins, Metrologist, on their diverse career paths spanning roles across academia, government agencies and industry and learn about their experience working at Sea-Bird...
The new HydroCAT-EP V2 provides some useful upgrades and tools for extending your deployment length. Watch our latest Sea-Bird University session to learn how to use, maintain, and process data from our latest multiparameter instrument. This is a recording of the...
Multiparameter systems can do more than just add more data points - when combined, a holistic dataset can provide valuable insight into environmental interactions. Watch our presentation to the Scripps Technical Forum on how multiparameter instruments can help us...