Consider this hypothetical situation: you find a loose cable with a 6-pin connector and a 4-pin connector - exactly what you need to connect an oxygen sensor to your CTD. However, just because the cable ends fit on the instrument connectors, you still don’t know if...
January 2021 Newsletter
Q&A: Temperature Mismatch
Although they should match perfectly, the two sensors rarely reported the same temperature value throughout their deployment. Furthermore, the 16plusV2 appeared to stop logging, although the batteries still had life upon recovery. Can you identify the problem?
Scripps Technical Forum | AC-S: The Telescope Into the Rainbow of Ocean Life
Scripps Technical Forum Time: Thursday, April 06, 2023 | 12:00 PM to 1:00 PM Location: Hydraulics Lab Mezzanine Sea-Bird Scientific is grateful for the opportunity to present at a Technical Forum to the Scripps Institute of Oceanography community. In this session,...
February 2023 Newsletter
Featured Posts Want your questions answered? Send in a request for a FAQ for a chance for it to be made into a new piece of content! show more Customer ServiceContact UsInformationCustomer ServiceContact UsInformation
Announcing the NEW LED AC-S
We are more than ecstatic to announce the release of the NEW LED AC-S Spectral Absorption and Attenuation Sensor. It is the first of its kind, and a standalone transmissometer in the market. The AC-S provides in-situ optical data of beam transmission and absorption...